M90.4 SEPARATION VESSEL
Separation vessels are determined to avoid contact of measured fluid with measuring device. These vessels contain liquid separating measured fluid from the fluid in the measuring device. The vessels can be used if the measured fluid is corrosive, highly viscous or if it can freeze, solidify or condensate in the impulse piping. Liquid separators have to be selected so as not to mix or otherwise react with the measured fluid.
Size of the condensation vessel is determined with respect to requirements of ISO 2186 as well as with respect to temperature and pressure of the steam in the main pipeline.
Limits of use and design
Separation vessel body is designed for working pressure of 11 MPa under temperatures 200 °C, 300 ° C, 400 °C or 10 MPa under 500 °C. Separation vessel is equipped with a shut-off valve at the outlet as a standard. The highest working pressure of the shut-off valve is determined by the type of used shut-off valve. On request separation vessel can be supplied without shut-off valve at the outlet, and supplied just with a weld-on socket for mounting or fittings.
Separation vessel body is made of stainless steel X6CrNiTi18-10 (1.4541). Welded-on pieces, sockets and plugs are made of stainless steel. Shut-off valve seat is made of stainless steel X6CrNiTi18-10 (1.4541) or X6CrNiMoTi17-12-2 (1.4571). Shut-off ball valve is made of stainless steel X2CrNiMo17-12-2 (AISI 316/1.4408). On request separation vessel can be equipped with shut-off valve made of different material.
Separation vessel body is equipped with sleeves ended by thread or preparation to a welded joint on diameter 20 mm, 21.3 mm or 14 mm – at the inlet in upper part (away from the primary device) and at the outlet side of the vessel body (towards the secondary device) In the upper part of the device there is a welded-on piece blinded by a stopper for filling the vessel with a liquid separator. On the side there is a welded-on piece blinded with a stopper for indication of the separation liquid level. In the lower part of the vessel there is a welded on piece ended by a thread or preparation for a welded joint on diameter 14 mm up to 20 mm for discharge valve connection. On request separation vessel can be equipped with a different process connection.
Way of fixing
Separation vessels should be installed at the same level and as close as possible to the pressure tapping. Space between the vessel, impulse piping and sensor has to be completely filled with liquid separator.